FAIRBANKS SCALES SCB R9000 DRIVER
Parts Page 23 Page 24 – Appendix I: Comments to this Manuals Your Name. IF you are using a remote display, check the main display first, if it’s OK then check the cable and plug connection on the remote First unplug, then plug in the AC adapter to reset the program. Page 4 Section 8: All equipment should be protected from direct sunlight.
|Date Added:||19 June 2006|
|File Size:||14.58 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
It uses default Windows drivers and is installed automatically upon first connection with a PC. Specifications Page 13 – Environment Page 14 – Section 6: Weight readings will be accurate as soon as the unit is powered on and set to zero. Because the scale uses special low-power circuitry, no warm-up time is required. IF you are using a remote fairbaks, check the main display first, if it’s OK then check the cable and plug connection on the remote First unplug, then plug in scalees AC adapter to reset the program.
Fairbanks Ultegra SCB-R9000-14U Service Manual
Remove the scale from the packing box and place on a flat surface where it will be used. Using the Scale 3. Don’t have an account?
Customer Care Page 11 Page 12 – Section 5: Page 11 Section 4: Page 17 Section 6: Page of 29 Go. Units may be ordered with a ball roller-deck surface.
Comments to this Manuals Your Name. Turn the unit upside down and remove all four feet from the scale platform 3.
If the corners are not within one 1 division of each other, conduct a good visual inspection to verify that there are fairbanms binds or interference with the platform and re test the scale. When the unit is powered, the liquid crystal display LCD will show the software part number and the revision followed by either “0. All equipment should be protected from direct sunlight.
Page 2 Every effort has been made to provide complete and accurate information in this manual. The maximum USB cable length is 5 meters unless a self powered hub is used to extend the cable length.
Ultegra Roller Top (SCB-R)
Page 15 Section 6: This manual also for: Sealing The Scale Section 6: Repair and Parts Replacement Page 4 Section 8: If the corners are still not within one 1 division of each other, identify the corner s with the error and replace the load cell in that corner. By pressing ‘print’ button you will print only current page. General Information Page 6 – Section 2: Verify no acb is on the platform.
Calibrate the scale Section 6 Note: Disconnect power to the scale 2. Page 23 Section 8: Do NOT adjust all 4 feet. A low power USB device draws less than mA. Corner Test Section 6: