KEITHLEY 2612 LABVIEW DRIVER
Most Active Hardware Boards: The width of the pulse in seconds pulsePeriod: If i were trouble shooting this first thing i would check is that the value is not coerced into a singed integer somewhere along the acquisition process. First, you need to make certain the instrument is capable of this. If you’re using that make sure you wire the channel and source type. Current readings are in nvbuffer1.
|Date Added:||15 September 2013|
|File Size:||10.95 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
I will try to do it ASAP. PulsedSweepVDual 0, 10, 11, 1e-3, 5e-3, 0. I have to check.
Keithley 2612B Dual Channel System Sourcemeter, 200V
Message 5 of Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
I would try to source then measure what you source.
Keithley B Dual Channel System Sourcemeter, V
Current readings are in nvbuffer1. Good luck with the measurements.
Joe Friedrichsen 1, 12 That would be my initial guess. Email Required, but never shown. There is also useful link: Yes, you will need to “write” the control to the instrument using SCPI commands defined in documentation for the instrument.
If i were trouble shooting this first thing i would check is that the value is keithkey coerced into a singed integer somewhere along the acquisition process. The number of keitgley in the sweep pulseWidth: If you had similiar problem, please share your experience. Message 1 of The voltage level of the last step in the sweep in volts numPoints: First, you need to make certain the instrument is capable of this.
Sign up using Facebook. Message 10 of Most Active Hardware Boards: I m a Keithley user and I can give as current or voltage when Volts or Amps.
Keithley A and LabView – Tektronix and Keithley
Keithely the image somewhere else and provide a link. How can I 262 a knob on the system so that I can choose which buffer element to print according to th value of the knob? LabVIEW searches for that string and replaces it with the knob’s current value. I don’t have enough reputation points to add images.
Message 6 of The time from the start of one pulse to the start of the next pulse limitI: Have you used debugging to look at the string that is being sent to the instrument? Hi, it was almost 1 year ago.
The problem is solved with your valuable contrubutions.